A hand-held meter used to test the performance of ESD floors in combination with a person per ANSI/ESD STM 97.2. and IEC Standard 61340-4-5.
The electrostatic charge of a person walking on an ESD safe floor with ESD safe footwear can be measured with the EFM51WT Walking Test Kit. This test provides measurement data on the actual charge on a person while walking which is used to determine the effectiveness of the floor/footwear grounding system. Tests ESD Floor
The EFM51 holds the PEAK value of the measured body voltage, as well as the LIVE static voltage measured on the operator’s body.
The kit includes the EFM51 static field meter, walking test adapter plate, hand probe and conductive carrying case. The EFM51 is a fully function field meter that in addition to the Walking Test, measures and detects electrostatic fields, surface potentials and discharge times (with optional charge plate accessory - p/n 7100.EFM51.CPS.SET.)
Below is the procedure for the EFM51 walking test.
Prepare the EFM51
- The EFM should be placed into the grounded foam base that is on a flat surface near the floor to be tested.
- Hold the function button until "change mode" appears. Press the button until "Max Value" appears (This is 6 button presses). This mode will becomes active in 2seconds.
Hold the probe in one hand and perform the walking test:
The operator walks on the floor section in a six-step pattern and comes to rest with both feet on the floor specimen in front of the EFM meter. The step pattern requires forward and backward steps and a cross over step:
Perform the test three times and record the results.
Reading the meter:
The upper display line indicates the maximum positive and negative body voltage generated on the person when walking.
The lower line indicates the minimum body voltage on the person.
Understanding the measurements:
Two sets of measurements are obtained during the procedure. The first measures voltage generation during the walking cycle and is defined as the Maximum Body Voltage. The second measures the voltage obtained at the end of the cycle with both feet on the floor specimen and is defined as the Minimum Body Voltage.
MRO Essentials ships via the following companies:
Quick Ship by MRO Essentials
At MROessentials.com, we are on a mission to delivery quality products to you quickly. That is our value proposition. We offer quick turnaround times for purchase orders and same day shipping. Place an order after 3:30pm Eastern? It will ship the next day.
All shipments are FOB FACTORY. Purchaser is therefore responsible for both shipping charges and damage to goods in transit. Note that freight charges are based on weight AND volume. Assembled items take up a greater shipping volume and are typically charged at a higher freight rate than items shipped flat. Purchaser must specify a freight carrier at time the order is placed or MRO Essentials will choose a carrier and prepay shipments. The freight cost of prepay shipments will be added to each invoice unless otherwise specified. NOTE: Request to ship UPS 1st, 2nd, or 3rd day is a method of shipment and does not imply any expediting of order processing by MRO Essentials. For non-UPS shipments, the order will be shipped freight collect or third party billing at purchaser’s option. Purchaser is responsible for and agrees to pay (1) charges disallowed or billed back to MRO Essentials by third parties when purchaser has specified third party billing and (2) charges billed to MRO Essentials for corrections made by the carrier including address corrections. Some carriers may not be available to be used by MRO Essentials. Purchaser will be notified if an alternate carrier is required.
MULTIPLE SHIPPING POINTS:
MRO Essentials products are manufactured in various locations. Shipping points may vary and may change without prior notice.
INSURING YOUR SHIPMENT:
You are responsible for your shipment once it leaves the F.O.B. point. We recommend that you insure all shipments. It is our policy to insure shipments unless otherwise specified. The buyer is responsible for uninsured lost or damaged shipments.